For linoleum and other types of heavy duty flooring materials.
Heavy flooring adhesives generally produce high variable and time-consuming viscosity testing results when tested with standard viscometers and accessories. However, the R/S+SST in a yield test with modulus calculation (slope of line leading up to yield stress) quickly (30 seconds) qualifies the product with consistently reproducible results, thereby providing an improved QC procedure.
* While a particular model/version may be used as an example in this method, any current or past model/version from the same series may also be used. Please consult a sales associate to discuss the most current instrumentation and software available.
Controlled by Shear Rate (CSR)
Four types of adhesives are evaluated in the above graph. Modulus determination was set to evaluate up to 40% of peak stress value.
The following values were reported:
The vane spindle geometry easily accommodates adhesives of varying consistency and produces a repeatable curve that allows for quick pass/fail test.